1 Bathy-2010P CHIRP Subbottom Profiler / Deep Water Echo Sounder Overview Operation Installation Maintenance Applications
2 B2010P Overview - Components Bathy-2010P Sonar Unit Bathy-2010P Server Unit Bathy 2010P LCD Display/Keyboard/Touchpad Drawer Assembly LPT-2 2KW Linear Power Transmitter Bathy2010P Manual Bathy-2010P Installation Software
3 Bathy 2010P Overview Safety The following are general safety precautions that are not related d to any specific procedures and therefore do not appear elsewhere in this publication. These are recommended precautions that personnel must understand and apply during many phases of operation and maintenance. KEEP AWAY FROM LIVE CIRCUITS- Operating personnel must at all times observe all safety regulations. Do not replace components or make adjustments inside the equipment with the high voltage supply turned on. Under certain conditions, dangerous potentials may exist when the power control is in the off position, on, due to charges retained by capacitors. To avoid casualties, always remove power and discharge and ground a circuit before touching it. DO NOT SERVICE OR ADJUST- Under no circumstances should any person reach into to enter the enclosure for the purposes of servicing or adjusting the equipment except in the presence of someone who is capable of rendering aid. RESUSCITATION- Personnel working with or near high voltages should be familiar with modern methods of resuscitation. WARNING- 120VAC is present in the Bathy 2010 Unit and Server. Thus, the risk r of electrical shock and serious injury is present. All precautions should be taken to insure no human contact is made to the unit while power is on. WARNING- 240VAC, 120VAC and 300VDC are all present in the LPT Unit. They are only accessible after the safety shield is removed. Thus, the risk of electrical shock and serious injury is present. All precautions should be taken to insure no human contact is made to the unit while power is on. CAUTION- The Bathy 2010P has many ESD (Electrostatic Sensitive Devices). Normal precautions involve the use of anti-static protection material and ground straps for test personnel. DO NOT SERVICE OR ADJUST RESUSCITATION WARNING WARNING CAUTION
4 Bathy 2010P Overview Safety Shipping and Handling - The Bathy 2010P units are delivered in a Portable Ruggedized 19 rack mount with front and back protective covers and optional roller base unit. Emergency Operation and Shutdown - Under any emergency, adverse, or abnormal ship conditions, the Bathy-2010P system electronics should be powered down. The system is powered down by placing the toggle switch on the front panel of the LPT-2 2 into the - OFF position.
5 Bathy 2010P Overview System Data Acquisition Data Storage Data Playback Post-Processing Processing / CHIRP Sub Bottom Profiler Software Interface Linear Power Transmitter Hardcopy Devices Transducers (Frequencies/Power)
6 Bathy 2010P Overview System Interconnect
7 Bathy 2010P Overview Equipment Specifications Units Depth Ranges Shift Range Zoom Range Zoom Modes Display Strata Resolution Depth Resolution Depth Accuracy Speed of Sound Geographic Position Printer Output Feet or Meters 15, 30, 60, 120, 240, 480, 800, 1500, 2400, 3000,6000, 15000, Feet 5, 10, 20, 40, 80, 150, 300, 500, 750, 1000, 2000, 5000, Meters Auto-ranging modes in all units Feet in 1 Foot increments Meters in 1 Meter increments 30, 60, 120, 240, 480, 900,1500 Feet 10, 20, 40, 80, 160, 320, 640 Meters Bottom Zoom, Bottom Lock Zoom, Marker Zoom, GUI Zoom (Playback Only) Normal Data, Zoom Data, Navigation, Depth, Command/Status, Color Control for Data: 4 Selections or Custom (User Input), Data Color Invert possible Up to 6 cm (for Subbottom applications only) 0.1 Feet, 0.01 Meters Meets or exceeds all current IHO CHIRP Sub Bottom Profiler requirements for single beam echo sounders with Survey Grade Transducers. User Selectable 1500 Meters/Second, 4800 Feet/Second; Adjustable in increments of 1 M/sec or 1 FT/sec NMEA 0183, GLL, GGA, RMC, VTG, VHW, HDT Selectable Baud Rates (RS-232): 4800, 9600, 19200, Centronics (Parallel Port) interface to syqwest, Inc. s TDU Series Thermal Printers, EPC and Isys Thermal Recorders
8 Bathy 2010P Overview Equipment Specifications Specifications (continued) Shallow Water Operation Transmit Rate Event Marks Data File Output Data File Playback Frequency Output Transmit Output Power Input Power Dimensions Weight Environmental < 0.5 Meters; frequency and bottom type dependant Up to 4 Hz, depth and operator mode dependant Periodic, External, and/or Manual (Periodic selectable in 1 minute intervals) Stores in industry standard SEGY format. Either Raw data or Processed data may be selected. Stores Depth, Navigation, and Graphic Data in ODC format (Proprietary) Normal and Zoom Data stored is Pixel data and can be played back and/or printed Files can be played back and/or printed at Normal or Fast-Forward speed, with Pause and GUI Zoom and a Playback Scroll Bar for ease of file playback. 3.5KHz - 350KHz on either Channel 2KW (Pulsed) (maximum), multiple Power Level selections 10-30VDC Nominal Power < 60 Watts (sensor unit). AC Option Available 12.5 W(31.75cm) x 14 L (25.5cm) x 4.65 H (11.81cm) Bathy 2010P Sensor Unit 19 (48.26cm) x 16 (40.64cm) x 5.19 (13.2cm) Bathy 2010P 15 LCD Display 11lbs (4.98kg) Bathy 2010P Sensor Unit 16lbs (7.25kg) Bathy 2010P 15 LCD Panel PC/ Display 0 to 50 degrees C Operating Temperature 0 to 95 non-condensing
9 Bathy 2010P Host Software Main Window Toolbar Function Key Controls Navigation/Depth Display Gain and Auto Controls Range, Zoom Range, and Shift Controls Data Acquisition / Playback Color Palette and Unit Controls File Capture Status Range Markers Mouse Depth/Position Fields
10 The Toolbar Buttons
11 Menu Bar File Menu Start/Stop Recording Open for Playback Capture Picture Recent Files Exit
12 Recording Data Confirmation Window ODC Data Recording Mode SEG-Y Y Data Recording Mode Save ODC file as path selection Save ODC file as path selection SEG-Y Y File Header Information Start Recording ODC Data Only
13 Edit Menu Insert Event Mark Insert Annotation Configure Acquisition Parameters Configure Serial Ports Configure Eventing Configure Thermal Printer Configure Gate Limits Configure Clutter Configure Draft Configure Frequency Configure Heave
14 Inserting Annotations / Event Annotation Window Marks Configure Eventing Event Timing Manual or Periodic Annotate Event Mark With Position, Speed, Depth etc
21 Configure Gate Limits Enable Bottom Gate Limits Shallow Limit Deep Limit Entered Limits will be rounded to one decimal place!!
22 Clutter Setup The Clutter Level setting allows the user to dynamically adjust the color palette to eliminate low level noise in areas where the water is contaminated. The AUTO setting is good for most applications. Manual Clutter settings from 1 to 9 can be selected to the users liking. Level 1 permits the lowest signal levels to be viewed and Level 9 permits only the strong to moderate reply levels to be viewed. The optimum values for most applications are in the range.
23 Configure Draft Allows the user to compensate all sounding data for transducer location and ship's draft. User entered value. The Transducer Draft value is determined by measuring the distance from the waterline to the face of the transducer. This measurement provides an offset which can be used to determine the water depth below the surface or the depth below the transducer.
24 Configure Heave For high resolution survey requirements the heave input is used to correct the digitized depth values and the real time sounding data for the vertical motion of the vessel. The instantaneous Heave value is displayed on the system status bar and is stored in the ODC file.
25 Configure Frequency Allows the user to select the frequencies used for each channel of operation. Please note that this function is usually factory configured at the time of order.
26 User Preferences Playback files continuously (restart at end of file) Time Display Selection User can select Displaying Local or UTC Time. Alarm Count which is a threshold on which if the bottom is lost for a user selectable number of pings the Bathy 2010P will then signal; bottom lost.
27 View Menu Toolbar Status Bar Enable GUI Zoom Passive Pinger CH. 1-1 Selection CH. 2-2 Selection Dual Normal (No Zoom) Bottom Zoom Bottom Lock Zoom Marker Zoom Diagnostic View
28 Zoom Modes Enable GUI Zoom This option allows the user to digitally scale bottom data from a playback file and can function as a manual zoom.
29 Display Modes There are 4 display modes available with Bathy-2010P software, including 3 Zoom modes, and 1 without any zoom. Normal Bottom Zoom Bottom Lock Zoom Marker Zoom
30 Bottom Zoom Mode When selected, this mode will split the viewing area in half. The T left side will be used to show zoom information, and the right side for normal bottom tom data. Bottom Zoom mode centers the zoom display around the current depth allowing you to follow it up and down the water column at a high resolution. on. In addition, the user can use the Zoom Range Control to zoom in or out. The Zoom Range value represents the zoom range in whichever units s are currently selected. When the zoom window moves up or down to track the bottom, it does so in Zoom Range / 2 increments.
31 Bottom Lock Zoom Mode This mode functions the same way as Bottom Zoom, however it does not show the bottom moving through the water column. Instead, it will lock the bottom to the upper portion of the zoom view so that the user may continuously monitor sub-bottom bottom information.
32 Marker Zoom Mode Like in the other modes, Marker Zoom provides a high-resolution zoomed view of the water column, however in this mode, the user can specify where in the water column the zoom range will begin. This is accomplished by dragging the marker zoom bar up or down the normal view scale bar to the point you want the zoom range to begin. The T start of the zoom range can be observed at the top of the zoom window while you drag the marker zoom bar, but the marker zoom isn t t set until the mouse button is released.
33 Help Menu This menu includes a software Help guide along with an About Bathy-2010P option. Clicking it will display a window with information such as the Bathy-2010P 2010P s s software version and SyQwest, Inc. s s company information. In addition, the software will request and display Hardware/Firmware version information from the Bathy-2010P 2010P s s DSP to ensure it is connected and communicating properly.
34 Theory of Operation The BATHY-2010P system is configured as a flexible acoustic measurement sensor device capable of both shallow/deep water hydrographic and subbottom profiling applications. Each of these applications depend on certain characteristics in order to perform accurately and reliably. For hydrographic applications, the BATHY-2010P is capable of providing sophisticated algorithms for peak signal detection, automatic modes of: receiver gain, bottom tracking, pulse length and power level controls greatly reduce the probability of inaccurate bottom detection/tracking. For sub-bottom profiling applications the BATHY-2010P is capable of providing high energy/wide bandwidth transmit waveforms and an advanced bottom triggered TVG processing algorithm to facilitate both maximum bottom penetration along with high resolution layer definition.
35 Bathy 2010P Installation The Bathy 2010P system installation is comprised of the following items Bathy 2010P Sonar Unit Bathy 2010P Server Unit Keyboard / Display Unit LPT-2 Amplifier Portable Rackmount Case Bathy 2010P Transducer(s) Assembly Bathy 2010P Cables (Power, Data, Transducer) Transducer Mounting Hardware 110/220 Volt AC Power Source
36 Bathy 2010P System Interconnection Diagram
37 Getting Started Unpacking and Inspection Use care when unpacking the unit from the shipping carton to prevent damage to the contents. It is also recommended that the carton and the interior packing material be saved even after the unit has been installed on the vessel. In the unlikely event that it is necessary to return the unit to the factory, the original carton and packing material should be used. Verify that all parts described in the next section have been shipped with the unit.
38 Bathy 2010P Accessory kit The following (Table 7-1) 7 is a list of the accessory kit supplied with the Bathy 2010P CHIRP Sub Bottom Profiler / Echo Sounder.
39 Transducers, Accessories To complete the Bathy 2010P CHIRP Sub Bottom Profiler system solution the user needs to select the desired transducer(s) for the intended application. The list below shows the most common transducers sold with the Bathy 2010P. Contact SyQwest with any questions or issues regarding the optimum transducer configuration for your application. The other listed items are available to complement and enhance the operation of the Bathy 2010P echosounder. Please contact your authorized SyQwest distributor or visit our web site for information and assistance in obtaining any of these items. and Options
40 Installing the Bathy 2010P Electronics Selecting a Location for the Electronics The Bathy 2010P Sensor Unit is designed for portable, marine applications but maybe used in permanent installations as well. The user must determine e if the Bathy 2010P Sensor Unit is to be mounted on the vessel or just placed in a convenient place on-board the vessel. Either way the appropriate location for the unit u needs to be determined. The following considerations should be investigated igated before deciding upon a location: 1. Display Unit - Select the optimum place where data may be viewed. 2. The Sonar Unit must be installed into a 110/220 VAC Power Source.. The unit must also provide adequate access for cabling termination without binding, and allow suitable space for servicing the equipment. (If necessary, the data cable may be extended beyond 10 feet.) 3. Water Resistance. - The unit is NOT watertight or splash proof. 4. Environmental. - The operating temperature is 25C to +60C.
41 Installing the Bathy 2010P General Transducer Transducer(s) The Bathy 2010P is offered equipped with a single frequency or dual d frequency transducer that is designed for portable, over-the the-side mount applications. It may also be used in permanent installations. Before installing the transducer, the installer should read and understand the appropriate section below to insure that all of the installation issues are considered. Selecting a Location for the Transducer The location of the transducer is very important for maintaining g reliable bottom tracking and optimum performance of the equipment. Avoid installing ling transducers in locations where the transducer will be subjected to turbulent water, air bubbles, or vibration. The best clear water location on most vessels meeting g these criteria is approximately 1/3 the length of the vessel, aft from the bow.
42 Selecting a Location for the Transducer The location of the transducer is very important for maintaining reliable bottom tracking and optimum performance of the equipment. Avoid installing transducers in locations where the transducer will be subjected to turbulent water, air bubbles, or vibration. The best clear water location on most vessels meeting these criteria is approximately 1/3 the length of the vessel, aft from the bow.
43 Handling Transducers The transducer is the heart of the Bathy 2010P system and, in spite of its appearance and size, is a delicate instrument. Although it is designed d to be in contact with and survive tough marine environments, it should not t be dropped or mishandled during the installation. Caution is advised when handling the transducer to prevent any damage to the transducer face or radiating surface. The transducer and mounting assembly should be as clean and smooth as possible so the path of the sounding energy is uninterrupted. The T transducer face must not be painted with lead based bottom paint. In portable applications the transducer and mounting assembly should be cleaned with fresh water after use. WARNING: Do not expose the transducer to any solvents when cleaning any excess sealants. Strong solvents may damage the face of the transducer. Also, when handling the transducer, avoid lifting or pulling on the transducer cable. Although the cable appears thick and substantial, the internal cable wiring could be damaged by stress from the sheer weight of the transducer t and cause a malfunction at the most inopportune time.
44 Over the side mounted transducer installation example: CAUTION: Never pull, carry or hold the transducer by the cable as this may sever internal connections. Over The Side Transducer Mounting (Example mount shown as reference only)
45 Portable Transducer Maintenance Aquatic growth can accumulate rapidly on the transducer's surface reducing its performance in weeks. Clean the surface, keeping it free of marine growth and petroleum residue, with a soft cloth and mild household detergent. Inspect the cable periodically for kinks, abrasions a and cuts. Repair any damage using an approved waterproofing cable repair system. Inspect connections for indications of corrosion. WARNING: NEVER USE SOLVENTS! Certain cleaners, gasoline, paint, sealants and other products may contain strong solvents, such as acetone, which can attack many m plastics dramatically reducing their strength. Clean surface of transducer with a mild detergent only.
46 Permanent Transducer Installation For some applications it may be necessary and/or convenient to mount the Bathy 2010P Transducer permanently in the hull of the vessel. Due to the construction of the Bathy 2010P Transducer it may be directly affixed to the hull of the vessel. A Seachest Installation tion can be used, but is not required. Guidelines for performing a Seachest installation are described in the following sections. Seachest Transducer Installation Interior Seachest installations are best suited for solid fiberglass hulls to permit a minimum attenuation of acoustic reply signals. Hulls of other type material types may m be considered but most other hull types will require that a Seachest design be built into the hull with an acoustic window across the face of the enclosure (i.e. a significant hole must be cut out of the hull). Inside mounting to the hull does minimize drag to allow faster survey s speeds, however, loss of performance may result due to the attenuation loss in the hull. Locate the transducer where the hull is solid fiberglass resin to maximize sound transmission. Do not locate over balsa wood core material. Consult the hull manufacturer if you are unsure of the core material or the best location. Never bond large resin housings directly d to the hull; always use a liquid- filled box. In any permanent installation the intended final configuration should s be tested before it is implemented, if possible. For more information regarding the installation of a Seachest or other permanent transducer mount, refer to our website at or contact us directly.
47 Bathy 2010P Host PC Electrical Connections Connecting a GPS or other Navigation input to the PC running the Bathy 2010P software allows the user to store and annotate Date, Time, Position, and Heading information to the Acoustic data returns. The Bathy 2010P PC Software supports the NMEA 0183 protocol on a COM port that is software selectable by the user. When selecting a PC to use with the Bathy 2010P system the user should ensure that PC hardware supports multiple COM Ports if a Navigation input is desired. The user should refer to the GPS NMEA 0183 output connection information in their GPS Manual as well as the PC COM port wiring information in their PC Manual to insure that the Navigation input is wired correctly. GPS Connection
48 Bathy 2010P Depth Data Output & External Event Connections The Bathy 2010P PC Host application software also supports serial l port connections for the Depth Data output and External Event input that are typical associated with interfacing to CHIRP Sub Bottom Profiler Hydrographic Survey and Post-Processing Processing Software packages (i.e. Hypack Max, HYDROpro, SonarWizMAP+SBP..). Printer Connection The Bathy 2010P PC Software allows the user to interface to the all of the SYQWEST TDU Thermal Printers. The connection is established through the PC s Parallel Printer Port (25-pin D-Type) D as shown below in Figure Once the software has been started, all displayed acoustic data can be printed to the TDU in either Acquisition or Playback mode. The standard TDU Printer cable is provided with the printer and is 12 feet long and does not require any user wiring. Installers must locate the printer accordingly. PC Printer Port
49 Bathy 2010P PC Software Installation This section describes how to install the Bathy 2010P PC software e package. It is assumed that the reader has a working knowledge of installing Windows 95/98/2000/NT/XP /98/2000/NT/XP software. The installation software is located on the CD-ROM disc included with your Bathy 2010P. NOTE* It is recommended that you exit all running applications before inserting the CD and beginning the installation. To install the PC software, insert the Bathy 2010P CD into an available ailable drive. If Auto Insert Notification is enabled on the CD-ROM drive, then the Bathy 2010P PC installer will begin automatically. If the CD does not auto-start, simply execute the SETUP.EXE file in the root directory of the CD. Once the installer is running, it will verify that your operating system is compatible with the Bathy 2010P software, and then it will check which version of the Windows Installer program is installed in your system. If the Windows Installer program is not found or out of date, it will update it and prompt you to restart your computer. Once restarted, the Bathy 2010P installation will continue automatically.
AUD-220 Installation Guide STEREO MONO BRIDGE IR RS232 TX RX MIC 48V LINE L R MIC 1 2 INPUTS 24V DC 1 x 40W @ 8Ω 2 x 20W @ 4Ω LOOP OUTPUTS The Intelix AUD-220 is a 2x20 watt Class D amplifier with 8Ω speaker
SAFETY GUIDE This manual contains the precautions necessary to ensure your personal safety as well as for protection for the products and the connected equipment. These precautions are highlighted with
ATK-2040 AC/DC TRMS Watt Clamp Meter Users Manual EN 61010-2-032 CAT II 600V, CAT III 300V Pollution Degree 2 SYMBOLS showed on the clamp meter or in this manual: Caution, risk of danger. Refer to accompanying
UWE-1707 SERIES CONTENTS ELECTRONIC WEIGHING INDICATOR OPERATION MANUAL 1. SPECIFICATIONS PLEASE READ THIS MANUAL VERY CAREFULLY BEFORE ATTEMPT TO OPERATE THE SCALE 2. INSTALLATION 3. ROUTINE OPERATION
and brief description of Marine Gravity Meter KSS 32- M Copyright 1999-2010 Bodensee Gravitymeter Geosystem GmbH All rights reserved 1 KSS32-M Marine Gravity Meter Cover removed Copyright 1999-2010 Bodensee
Preface Obtaining Documentation and Submitting a Service Request, page xvii Safety s, page xvii Safety Guidelines, page xx Obtaining Documentation and Submitting a Service Request For information on obtaining
1550 1-36V, 0-3A DC Power Supply User Manual Safety Summary The following safety precautions apply to both operating and maintenance personnel and must be followed during all phases of operation, service,
User Guide Extech EX205T True RMS Digital Multimeter Extech EX210T True RMS Digital Multimeter IR True RMS Multimeter Extech EX205T Introduction Thank you for selecting the Extech EX205T True RMS Auto-ranging
Model 380950 80A Mini AC/DC Clamp Meter User's Guide Introduction Congratulations on your purchase of the Extech 80A Mini AC/DC Clamp Meter. The Model 380950 measures AC/DC Current, AC/DC Voltage, Resistance,
User s Guide 600A True RMS AC/DC Clamp Meter Model 38389 Safety International Safety Symbols This symbol, adjacent to another symbol or terminal, indicates the user must refer to the manual for further
Rhino Buffer Module PSM24-BFM600S Operating Instructions RHINO BUFFER MODULE PSM24-BFM600S Description The PSM24-BFM600S Buffer Module will hold the output voltage of a 24 VDC power supply after brownouts
Peak G 15min Handheld Electrostatic Meter Series IZH10 How to Order IZH10 Option Nil H None High-voltage measuring handle Accessories and Option / Part Number for Individual Parts The ground wire and soft
User Manual TL-DA14-HD2 1x4 HDMI Splitter & Distribution Amplifier All Rights Reserved Version: TL-DA14-HD2_160926 Preface Read this user manual carefully before using this product. Pictures shown in this
PRO1250D CT MID DIN rail three phase four wire energy meter. User manual Version 1.5 1 Safety instructions...3 2 Foreword...4 3 CE certificates...5 4 MID certificate...7 5 Performance criteria...8 6 Specifications...8
PRO1250D CT M-bus MID DIN rail three phase four wire energy meter. User manual Version 1.11 1 Safety instructions...3 2 Foreword...4 3 MID certificate...5 4 CE certificates...6 6 Performance criteria...8
USER MANUAL DV-HAS HDMI Audio Separator 244 Bergen Boulevard, Woodland Park, NJ 07424 Tel 973-785-4347 FAX 973-785-3318 Web: www.fsrinc.com LIT1460 Proprietary Information All information in this manual
EOS-6000 Series Optical A/B Switch User Manual DC Version For more information on this and other products: Contact Sales at EMCORE 626-293-3400, or visit www.emcore.com. Table of Contents Table of Contents...2
Keithley Instruments 28775 Aurora Road Cleveland, Ohio 44139 1-800-935-5595 http://www.keithley.com Model 8020-STC Kelvin Standard Triaxial Connector Card Description The Model 8020-STC Kelvin Standard
Keithley Instruments, Inc. 28775 Aurora Road Cleveland, Ohio 44139 1-888-KEITHLEY http://www.keithley.com Model 2657A-LIM-3 LO Interconnect Module User's Guide Description The Model 2657A-LIM-3 LO Interconnect
INSTRUCTION MANUAL TRIP CIRCUIT SUPERVISION RELAY GKAD1 TOSHIBA Corporation 2004 All Rights Reserved. ( Ver. 1.6 ) Safety Precautions Before using this product, please read this chapter carefully. This
Model: 601B SLA Battery Capacity Analyzer USER MANUAL Safety Summary The following safety precautions apply to both operating and maintenance personnel and must be followed during all phases of operation,
MulTI-Touch Projected Capacitive Touch Screen Integration Guide Document Number: 6500493 Rev. 1.1 Pg. 1 The information provided in this document is intended as a guide only and is subject to change without
3-4 SAS/SATA II HDD Canister Entry version XC-34D1-SA10-0-R USER S MANUAL Document number: MAN-00077-A ii Preface Important Information Warranty Our product is warranted against defects in materials and
Multi-Function Displays (A3416 & A3417) - Overview, Configuration MULTI-FUNCTION DISPLAYS A3416: 8 INCH / A3417: 13.3 INCH The OctoPlex A3416/A3417, Multi-Function Displays allow for the users to control
User's Guide High Precision Quad Output DC Power Supply Model 382270 Introduction Thank you for selecting the Extech Model 382270. This device is shipped fully tested and calibrated and, with proper use,
This Datasheet for the IC660BBA104 Block 115Vac/125Vdc Analog Current Source 4 Inputs / 2 Outputs http://www.qualitrol.com/shop/p-14425-ic660bba104.aspx Provides the wiring diagrams and installation guidelines
MODEL INSTALLATION INSTRUCTIONS FOR FEDERAL MODEL SELECTABLE SIRENTONE-MULTI VOLT INSTALLATION INSTRUCTIONS FOR FEDERAL MODEL SELECTABLE SIRENTONE-MULTI VOLT A. GENERAL. The Federal Model is a solid-state
Resolver to Digital Expansion Board Catalog No. EXB009A01 Installation and Operating Manual 6/98 MN1313 Table of Contents Section 1 General Information............................. 1-1 Introduction....................................
icore Kiosk system Installation Guide The reproduction, transmission or use of this document or its contents is not permitted without express authority. Offenders will be liable for damages. All rights,
A3.2 CR DUAL MONO POWER AMPLIFIER C R MUSICAL FIDELITY A3.2 DUAL MONO POWER AMPLIFIER POWER INSTRUCTIONS FOR USE Thank you for purchasing the Musical Fidelity A3.2 CR Power Amplifier. Used properly and
Manual Version: V1.00 Video Decoder User Manual Thank you for purchasing our product. If there are any questions, or requests, please do not hesitate to contact the dealer. Copyright Copyright 2016 Zhejiang
Owner's Manual True RMS Multimeter Model No. 82023 CAUTION: Read, understand and follow Safety Rules and Operating Instructions in this manual before using this product. Safety Operation Maintenance Español
Toll Free: 1-888-865-6888 Tel: 510-226-8368 Fax: 510-226-8968 Email: firstname.lastname@example.org User Manual LCDK 1070 DVI-D KVM Legal Information First English printing, October 2002 Information in this document
Dell MD1280 Storage Enclosure Getting Started Guide Regulatory Model: SP-2584, E11J Notes, Cautions, and Warnings NOTE: A NOTE indicates important information that helps you make better use of your computer.
The InoxLaser uses time-offlight laser technology for continuous non-contact distance or level measurement. Able to measure to any solid surface, at any angle, the an ideal sensor for level, positioning
User's Guide 99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com Energy Logger Model EM100 Introduction Congratulations on your purchase
Model 820-218 RD-4 Remote Time Display Serial Number March 28, 1997 Revision D TABLE OF CONTENTS SECTION PARAGRAPH TITLE ONE GENERAL INFORMATION 1.1 Scope of Manual 1.2 Purpose of Equipment 1.3 Physical
ELECTROSTATIC FIELDMETER Small hand-held Electro-Fieldmeter with digital display designed to measure electrostatic voltage potentials (with pre-selected distance) according the fieldmill influence principal.
CyberView User Manual RKP215-801 RKP217-801 RKP215-1601 RKP217-1601 2U Rackmount LCD monitor Keyboard drawer with KVM switch Version 2.0 20 March 2004 Table of Contents 1.0 General... 3 1.1 Unit Introduction...
VDI Pro Voltage & Dry Contact Interface Installation and programming Guide MODEL VDI MK2 1 of 18 PREFACE Important Installation Information It is the purchasers responsibility to determine the suitability
1/32-DIN TEMPERATURE CONTROLLER INSTALLATION, WIRING AND OPERATION MANUAL FORM 3882 This manual is intended for use in support of installation, commissioning and configuration of the 1/32-DIN Temperature
Model: Pro93 TRUE RMS LEAKAGE CURRENT TESTER CONTENTS TITLE PAGE I. Safety Information.......1 Environmental Conditions... 1 Explanation of Symbols..... 1 II. Specification..... 2 General Specification...
DWP-PC SERIES PRICE COMPUTING SCALE User s Guide PLEASE READ THIS MANUAL VERY CAREFULLY BEFORE ATTEMPT TO OPERATE THE INSTRUMENT Specifications subject to change without prior notice CONTENTS 1. INSTALLATION
User's Guide Model 382270 High Precision Quad Output DC Power Supply Introduction Congratulations on your purchase of the Extech 382270 DC Power Supply. The Model 382270 can be used for many applications
OWNER S MANUAL SINGLE CAT5E/6 3D EXTENDER B-200-1CAT-HDIR Receiver IMPORTANT SAFETY INSTRUCTIONS WARNING: To reduce the risk of fire or electric shock, do not expose this apparatus to rain or moisture.
NTP-5521/5531/5561 SWITCHING MODE POWER SUPPLY USER MANUAL Keep this manual in a safe place for quick reference at all times. This manual contains important safety and operation instructions for correct
HD IR Vari-Focal Fixed Network Camera Quick Start Guide Model No. K-EW114L01E Version 1.0.0 Welcome Thank you for purchasing our network camera! This quick start guide is designed to be a reference tool
5 B&W Rear View System Camera Instruction Manual MODEL: CA453 www.lorexcctv.com Copyright 2007 LOREX Technology Inc. Thank you for purchasing the Lorex 5 Black & White Rear View System Camera. This system
OPERATING INSTRUCTION AUTORANGING MULTIMETER MAX Ω F C 10A MAX every 15 min. COM V SAFETY INFORMATION The following safety information must be observed to insure maximum personal safety during the operation
QUICK START GUIDE KEPCO An ISO 9001 Company. ATE VOLTAGE/CURRENT STABILIZED POWER SUPPLY I INTRODUCTION 1.1. SCOPE OF MANUAL. This Quick Start Guide covers simple installation and local operation of the
BCM2 Series Branch Circuit Monitors Quick Setup Guide Safety Information DANGER! HAZARD OF ELECTRIC SHOCK, EXPLOSION, OR ARC FLASH Follow safe electrical work practices. See NFPA 70E in the USA, or applicable
1993 Specifications CSJ 0598-01-057 1.0 Description SPECIAL SPECIFICATION ITEM 6398 Transportation Management Console This Item shall govern for the furnishing and installation of Transportation Management
C O N T E N T S TITLE PAGE I. Safety Information. 1 II. General Description.. 1 III. Specifications.. 2-3 IV. Controls... 4-5 V. Calibration Procedures........ 6 VI. Measurement Preparation....7 VII. Operating
500424 MuxLab Inc. 2016 94-000808-A / SE-000808-A SAFETY PRECAUTIONS To insure the best from the product, please read all instructions carefully before using the device. Save this manual for further reference.
Agilent Technologies E5346A 38-Pin Probe and E5351A 38-Pin Adapter Cable Installation Note The 38-pin probe and adapter cable provide a convenient way to connect two Agilent Technologies logic analyzer
Vision OPLC V130-33-T38 Installation Guide The Unitronics V130-33-T38 offers the following onboard I/Os: 22 Digital Inputs, configurable via wiring to include 2 Analog and 2 HSC/Shaft-encoder Inputs 16
FACTORY AUTOMATION QUICK START GUIDE ODT-HH-MAH300 Handheld Laser Class 2M EN 60825-1 The General Terms of Delivery for Products and Services of the Electrical Industry, published by the Central Association
For Research Use Only. Not for use in diagnostic procedures. Thank you for purchasing the Agilent G5590A BenchCel Workstation. To get you started and to assure a successful and timely installation, please
JR3 EXTERNAL SENSOR ELECTRONICS WITH SERIAL DATA OUTPUT JR3, Inc. 22 Harter Ave. Woodland, CA 95776 5963B 13 October, 2003 TABLE OF CONTENTS CHAPTER 1 OVERVIEW General...1-1 Sensor...1-1 Electronic System...1-1
PRO1 V21.xxx series MID DIN rail single phase two wire energy meter PRO1D V21.212 Version 1.1 Note: the picture on the frontpage is a meter from the series of this meter, it might not be exactly the same
SIMATIC VPU200 3104 Hardware Manual Order Number: PPX:560/65 8102 Manual Assembly Number: 2597773 0006 Second Edition ! DANGER DANGER indicates an imminently hazardous situation that, if not avoided, will
Programmable Voltage Interface Unit Models VSP12824V & VSP12860V Installation & Programming Manual PREFACE Important Installation Information It is the purchasers responsibility to determine the suitability
USER S MANUAL Data Logger SOUND LEVEL METER www.galaxyaudio.com/products/cm-170 This page intentionally left blank Contents Introduction...1 Safety Information...2 Overview...3-4 Before Getting Started...5
STRIKER PLUS 4/5/7/9 Owner s Manual 2017 Garmin Ltd. or its subsidiaries All rights reserved. Under the copyright laws, this manual may not be copied, in whole or in part, without the written consent of
Inertial Measurement Unit (IMU) ISIS-IMU (Rev. C) Features Fully compensated Inertial Measurement Unit (IMU) DC in digital output Most cost effective IMU - Lowest cost 6 degree of freedom IMU in its performance
MBB300C Vessel Monitoring and Control Black Box User s Manual Revision 1.0 Copyright 2016 Maretron, LLP All Rights Reserved Maretron, LLP 9014 N. 23 rd Ave #10 Phoenix, AZ 85021-7850 http://www.maretron.com
SuperLoader 3 Contents SuperLoader 3 Quick Start Guide The Quantum SuperLoader 3 is an automated storage and retrieval device consisting of a tape drive and up to two tape magazines holding up to eight
PM Series Power Meter Quick Setup Guide - PMC-1000, PMC- 1001, PMM-1000, PMB-1960 Safety Information DANGER! HAZARD OF ELECTRIC SHOCK, EXPLOSION, OR ARC FLASH Follow safe electrical work practices. See